# Category Archives: Theory and tutorials

## Some ways to measure ESR

There are many misunderstandings about measuring electrolytic caps and their ESR. 99% of the ESR meters are hobby instruments and more indicators as meters. However they can be usable if you understand how to use them and what they measure. … Continue reading

Posted in Theory about measurements | Comments Off on Some ways to measure ESR

## Teardown and trouble shooting an anti-mice beeper

This is a dead anti mice thing you plug in a mains outlet. I had a dead one and was curious what was inside. The problem was a dead cap and some burned contacts. I wanted to know the kind … Continue reading

| Comments Off on Teardown and trouble shooting an anti-mice beeper

## Short circuits finding

If you have a short in a circuit there are several ways to find them. A short can be a bridge between 2 traces (solderblob or manufacturing fault) , a component (like a short cap) or a protection device that … Continue reading

| Comments Off on Short circuits finding

## How to repair testgear (and other electronics)

A lot of the measurement stuff I have, came to me in non working, or long not used condition. Repairing is something I enjoy very much. You learn a lot from it and if you succeed it feels good. A … Continue reading

| Comments Off on How to repair testgear (and other electronics)

## Measuring inductor-cores and transformers

OK, you say, I have a LCR meter or a bridge so that’s a piece of cake. But inductors are kind a tricky when it comes to measuring. Especcially when they use core material. When I talk about inductors on … Continue reading

| Comments Off on Measuring inductor-cores and transformers

## LCR meter tutorial

An LCR meter is a typername for an instrument that is capable of measuring: Selfinductance of inductors and transformers Capacitance Resistance Some can do more, like conductance, the inverse of resistance. (G in mho or Siemens). History First the base. … Continue reading

Posted in theory about instruments | Comments Off on LCR meter tutorial

## Bridge measurements 1, capacitance

As you can see on my site I have a few LCR bridges and they are great to measure LCR. Most are very accurate. % is not always looking great compared to modern LCR meters but they are real “worst … Continue reading

Posted in Antennas and tuners, Theory and tutorials | Comments Off on Bridge measurements 1, capacitance

## About AC and RMS meters

There are several ways to note the amplitude of an AC signal. An AC signal has a value that switches over a certain time from V- to V+. It alternates between those values. The way to note this value is as … Continue reading

Posted in Multimeters, Theory about measurements | Comments Off on About AC and RMS meters

## All about ESR measurement

ESR is not new. It is not magical, it is not complicated. So it should be easy to measure. Just buy one of the those 50-100 Dollar wonderboxes and you can measure it. But do you now what you are … Continue reading

| Comments Off on All about ESR measurement

## Het electronen stroom broodje aap verhaal.

Stroom is de verplaatsing van electronen door een geleider , toch?  Dat leren we tenminste. Maar zo is het niet. De verwarring begint in de tijd van Maxwell. Omdat electronen nog niet bekend waren (pas in 1897 door Thomson) dacht … Continue reading

Posted in Theory general electronics | Comments Off on Het electronen stroom broodje aap verhaal.

## Good electronic Books

Knowledge does not come by itself, beside internet there are curious things named books. They are a good source of knowledge. This is a part of my collection. The concepts range from Tektronix, made in the 60’s. On the picture … Continue reading

Posted in Other stuff, Theory general electronics | Comments Off on Good electronic Books

## (bijna) alles over multimeters

multimetertutorial Dit is de downloadbare pdf. Multimeter Tutorial.                 Onmisbaar in onze hobby, maar als je er een zoekt dan is het aanbod enorm en het prijsverschil nog groter. Je koopt ze nieuwe … Continue reading

Posted in Measurement gear, theory about instruments | Comments Off on (bijna) alles over multimeters

## Multimeter specs vergelijken

Er zijn multimeters van minder dan 1 euro tot over de 1000 euro te koop. Als je de specs bekijkt zie je vaak met het stijgen van de prijs steeds meer getallen. Maar wat zeggen die getallen nu. DIGITS, dat … Continue reading

Posted in theory about instruments | Comments Off on Multimeter specs vergelijken

## Humor

Posted in Theory and tutorials | Comments Off on Humor

## meting aan trifilaire 1:9 unun

Dit is het principe van de unun, meer over baluns/ununs hier: http://www.pa4tim.nl/?p=52 Een 1:9 zet de impedantie van je antenne om naar een 9x lagere.  Dat doet hij met het ohmse maar ook het imaginaire deel.  Maar dat doet hij … Continue reading

| Comments Off on meting aan trifilaire 1:9 unun

## Invloed van de meetopstelling bij meten aan kringen

Als je aan een kring meet zul je altijd die kring beinvloeden. Wat je ook doet. Daarom is er een belaste en een onbelaste Q. Bij resonantie geeft in theorie de condensator al zijn energie aan de spoel en de … Continue reading

| Comments Off on Invloed van de meetopstelling bij meten aan kringen

## Redenen om zelf te repareren

Posted in Theory about measurements | Comments Off on Redenen om zelf te repareren

## Condensators meten zonder LCR

Ook zonder LCR meter kan de de waarde van een condensator bepalen. De meest bekende manier is natuurlijk om er een bekende spoel aan te hangen en dan de resonantie frequentie berekenen maar het kan simpeler. Een condensator heeft een … Continue reading

Posted in theory about components | Comments Off on Condensators meten zonder LCR

## Network analysis tutorial

Deze cursus is in het Engels en Frans gemaakt. Chapter 1 is about the influence of your dut holder and calibration. Chapter 2 is about the behaviour of inductors. Chapter 3 is about capacitance. Chapter 4 is about resonance and … Continue reading

Posted in Network analyse tutorials | Comments Off on Network analysis tutorial

## DC leakage and reforming electrolytic caps

This page is one of my pages about electrolytic caps. In this case DC leakage. Instead of just replacing caps to be sure, I wanted to measure the DC leakage and for most be able to reform them. Electrolytics have … Continue reading

| Comments Off on DC leakage and reforming electrolytic caps

## Time Domain Reflection meting (TDR)

Time domain reflectie meting is een erg handig en makkelijk principe. Grof gezegd, je zet een puls op een transmissielijn. De scoop hangt via een T stuk samen met de pulsgenerator aan dat stuk coax. Er gaat een puls uit … Continue reading

| Comments Off on Time Domain Reflection meting (TDR)

## HP cross-reference eigen partnummers-industrie nummers

Deze lijst bevat veel eigen HP component nummers met daarachter het JEDEC nummer. Best handig om te weten dat dat obsolete HP torretje met de benaming 1854-0210 gewoon een 2n2222 is  Met dank aan de onbekende samensteller. Tweede lijst gaat … Continue reading

Posted in Theory about measurements | Comments Off on HP cross-reference eigen partnummers-industrie nummers

## Tek boeken

Tek heeft in de jaren 60 een serie paperbacks uitgegeven over het ontwerp van meetaaparatuur. Dit zijn ze niet allemaal. Ik heb er meer uit deze serie, een boek over probe measurements en een over curve trace metingen.

| Comments Off on Tek boeken

## Hoe betrouwbaar is je meter bij niet perfecte spanningen

Een multimeter heeft een aantal specificaties en er zijn diverse soorten. Belangrijk is het dus om je meter te kennen. Laten we beginnen met wisselspanning. Een wisselspanning heeft een belangrijk kenmerk. De amplitude is niet constant. Maar hij hoeft niet … Continue reading

| Comments Off on Hoe betrouwbaar is je meter bij niet perfecte spanningen

## Scope Bandwith measurements

A scope has a defined bandwidth. If you have a 100Mhz scope this means the amplitude from your signal is down 3dB at 100MHz. This does not seem much but it is 0,707 times the real value. 6dB is only … Continue reading

| Comments Off on Scope Bandwith measurements

## ESR meten (also in English)

An updated version: http://www.pa4tim.nl/?p=3775 Een ELCO bestaat in werkelijkheid uit een weerstand (ESR), spoel (ESL) en de eigenlijke ELCO. Hier gaan we proberen deze zaken te meten. Om dat te doen zijn er diverse methodes. De beste manieren zijn speciale vectoriale … Continue reading

| Comments Off on ESR meten (also in English)

## Weerstanden zijn niet altijd wat ze lijken

Iedereen weet dat sommige weerstanden voor HF meer of minder geschikt zijn maar wat zijn nu precies de effecten en eigenschappen. Koolweerstanden kunnen tot hoge frequenties gebruikt worden. Hou wel de pootjes zo kort mogelijk. Metaalfilm is helemaal mooi. Draadgewonden … Continue reading

| Comments Off on Weerstanden zijn niet altijd wat ze lijken

## Condensators

We weten allemaal dat een condensator een reactantie heeft. Neemt de frequentie toe dan neemt de reactantie af. Dat is wiskundig te berekenen. Maar wat is er nu mooier dan het gewoon zichtbaar te maken. Hoe lager de trace hoe … Continue reading

| Comments Off on Condensators

## Advies over scopen ivm aanschaf

In deze pdf vind je informatie over de verschillende soorten scopen, dingen waar op je moet letten bij tweedehands en nieuw aankoop en uitleg over de specificaties en accesoires. Download PDF

| Comments Off on Advies over scopen ivm aanschaf

## Replacing Electrolytic caps …or not ?

Electrolytic capacitors (=caps in this text) are parts that often cause problems. So worth the troubles to check them while doing a restoration.  The general opinion is replacing all caps.  Often not a bad idea but also a bit over … Continue reading

Posted in theory about components | Comments Off on Replacing Electrolytic caps …or not ?

## Uitleg over netwerkanalyse en richtkoppelaars

Hier ook het artikel met plaatjes en foto’s in een word document: reflecties_klaar.doc Staande golven iets geheimzinnigs, ik dacht het niet…. Hoofdstuk 1, een stukje noodzakelijke doch lichte theorie . Inleiding: In onze hobby hebben we veel te maken met … Continue reading

Posted in Network analyse tutorials | Comments Off on Uitleg over netwerkanalyse en richtkoppelaars

## Antenne tuners (also in English)

Al weer een stuk over tuners. Ach ja, waarom niet. Alleen nu geen kant en klaar recept maar uitleg over wat zo’n ding doet. Dat is namelijk best handig om te weten als je er een wil maken. Another peace … Continue reading

Posted in Theory and tutorials | Comments Off on Antenne tuners (also in English)

## SINAD, vergelijkende factor voor ontvangers

SINAD is de ratio van de totale power aangeboden, dus signaal, ruis en vervorming ten opzichte van het ongewilde signaal wat bestaat uit ruis en vervorming. Hoe hoger de SINAD hoe beter de ontvanger. Dit gaat vooral om de mate … Continue reading

Posted in Theory general electronics | Comments Off on SINAD, vergelijkende factor voor ontvangers

## dBm naar Volt naar Watt conversie tabel

Hier een tabel met hoge resolutie om deze waarden zo om te zetten. Dit is een voorbeeldje van hoe het er uit ziet. Wil je hem helemaal dan staat hier het bestand ook (onder de tabel). dm_watt_volt_tabel.xls

| Comments Off on dBm naar Volt naar Watt conversie tabel

## Ruis in ontvangers

Er zijn veel soorten ruis maar ik wil het hier hebben over thermische ruis. Thermische ruis is niets anders dan de “herrie” die electronen maken als ze in een substantie tegen hindernissen aan botsen. Ze ondervinden dus weerstand. Bij het … Continue reading

| Comments Off on Ruis in ontvangers

## Transistor bias

Hier wat plaatjes en formules waar je handig zelf mee aan torren kan rekenen. Het is veel makkelijker dan je denkt. Er zijn drie basis schakelingen waarvan de common emitter het meest gebruikt wordt in onze zelfbouw. Hoe pak je … Continue reading

Posted in Theory general electronics | Comments Off on Transistor bias

## Handige formules en feitjes die je altijd kwijt bent

Hier komen alleen formules (nog) geen uitleg over hoe en wat. Radialen per seconde: 2.PI.f (f in Herz) (Pi = 3,1415927) Reactantie: Xl=2.PI.f.L Xc= 1/2.PI.f.C in standaard waarden of 1000000/2.PI.f.C waarbij f in MHz en C in pF resonantie in … Continue reading

Posted in Theory general electronics | Comments Off on Handige formules en feitjes die je altijd kwijt bent

## Het makkelijk zelf berekenen van een filter

Dit Stukje tekst komt van FET, een wandelende RF computer op Circuits Online. Een heel heldere uitleg over het zelf berekenen van filters. Laten we een banddoorlaatfilter ontwerpen voor achter een FM zendertje. We laten hem lopen van 88 tot … Continue reading

Posted in Theory general electronics | Comments Off on Het makkelijk zelf berekenen van een filter

## Balun zin en onzin (also in English)

English version (short) below Er worden de meest wilde verhalen over baluns geschreven. Daarnaast wordt de balun vrolijk op een hoop gegooid met de unun. Een balun is een afkorting voor balanced-unbalanced. De unun is dus unbalanced naar unbalanced. Sommigen … Continue reading

Posted in Theory about ...., Theory and tutorials | Comments Off on Balun zin en onzin (also in English)

## Resonantie het belangrijkste bij antennes, denk het niet…

Hoe en antenne werkt heb ik al elders beschreven. Veel mensen zweren bij een antenne die resonant is. Leuk, maar het zegt niets. Ten eerste is er serie en parallel resonantie. De eerste is laagohmig. De tweede hoogohmig. De eerste … Continue reading

Posted in Theory about ...., Theory and tutorials | Comments Off on Resonantie het belangrijkste bij antennes, denk het niet…

## Reflecties, waaruit bestaan ze eigenlijk ?

To reflect or not to reflect, that’s the quest. Dit is een diepgaander stukje over reflecties. Een andere manier om er naar te kijken. We hebben allemaal geleerd dat wanneer R-in niet gelijk is aan R-uit er in een transmissielijn … Continue reading

| Comments Off on Reflecties, waaruit bestaan ze eigenlijk ?

## SWR: Wat verder gaande antenne theorie mbt aanpassing

SWR, als er ergens veel misverstanden over zijn is het over de SWR. De hoofdverwarring ontstaat doordat een meetzender wel 50 ohm is. Hij heeft een interne 50 ohm weerstand. Hij gaat dus niet stuk bij rare belastingen. Is de load … Continue reading

| Comments Off on SWR: Wat verder gaande antenne theorie mbt aanpassing

## Antennes de meestal overgeslagen basistheorie

Antennes, eigenlijk niet zo moeilijk, Deze tekst kan je ook downloaden (klik). Deze bevat dan ook wat tekeningen. Tja Antennes is niet eenvoudig lijkt het maar eigenlijk ook niet moeilijk. Een heel grof vergelijk: Zie een antenne als een gloeilamp. … Continue reading

| Comments Off on Antennes de meestal overgeslagen basistheorie